Replacement Conductivity Meter Sensor Head for the Horiba Conductivity Meter in the Bresle Patch Test. When replacing the head, it is important to calibrate as follows: Place 3 to 4 droplets of the 1413µS/cm Conductivity Solution into the measuring electrode, ensuring that the solution is in both sections of the electrode with no air bubbles. Check the displayed reading which is shown when the smiley face comes on and if this is not 1413µS/cm then calibrate as follows: Press the Cal button, the CAL indicator and a smiley face will flash – the Conductivity Meter will now auto calibrate. When the CAL indicator and smiley face stop flashing the calibration is complete and the instrument will revert to normal measurement mode. When you have finished calibrating the Conductivity Meter, the measuring electrode should be rinsed using the syringe with Deionised Water and shaken dry All other consumables for the Bresle Patch Test are available separately.
Replacement Conductivity Meter Sensor Head
Replacement Conductivity Meter Sensor Head for the Horiba Conductivity Meter in the Bresle Patch Test.
Replacement Conductivity Meter Sensor Head - Ref: T01-30226